- The instrument is designed and made as per T 0623 Bitumen Saybolt
Viscosity Test (Saybolt Viscometer Method) in the JTJ052 Standard
Test Methods of Bitumen and Bituminous Mixtures for Highway
- It is suitable to determine the Saybolt viscosity of bitumen and
bituminous mixtures under a constant temperature as per T0623.
II---Main technical features
1. This instrument is composed of bath, temperature controller,
parameters display and calculation(calibration coefficient, time,
viscosity and so on).
2. This instrument adopts micro computer technology.It equips
double lines to detect.It can determine two samples at the same
time. It can display the parameters and calculate the arithmetic
mean value automatically.
3. This instrument can convert the Saybolt viscosity to Engler
viscosity and kinematic viscosity automatically. Operator can get
three viscosity data from one test. The work efficiency is high.
III---Main technical specifications
1. Power supply: AC(220±10%)V, 50Hz
2. Instrument structure: Desk type
3. Working mode: Double lines,parallel test
4. Receiving flask: (60±0.0)5ml
5. Heating power of bath: 1000W
6. Working temperature of bath: Room temp.∼240.0°C
7. Temperature control accuracy: ±0.1°C
8. Timing range: 0.0s∼999.9s
9. Timing accuracy: ±0.1s
10. Ambient temperature: ≤35°C
11. Relative humidity: ≤85%
12. Dimension: 360mm×360mm×790mm